FX100 오디오 분석 시스템
XL2 사운드 레벨 미터
MR-PRO / MR2 제너레이터
TalkBox 음향 생성기
DR2 디지털 제너레이터
DL1 디지털 오디오 분석기
라이브 사운드(Live Sound)
동영상 모음 자료
FAQ (자주 묻는 질문)
My NTi Audio
Microphone Test Configurations (ko-KR)
Microphone Test Configurations
The microphone measurement system offers full quality control and parameter verification of
Build-in microphones in mobile phones or headsets
Setup with 1x Microphone
This setup verifies single microphones in a basic test environment. The sound source is equalized for a flat frequency response using a reference measurement microphone.
Calibrate and equalize the sound source by using the reference measurement microphone M2010
Start microphone measurements with the equalized sound source
Setup with 1x Microphone and 1x Reference Microphone
This setup verifies single microphones in a test environment with equalized sound source and reference microphone. The measurement provides results relative to the reference measurement microphone, thus the results are immune to any changes in the environment such as temperature or humidity.
Calibrate and equalize the sound source by using the permanently-installed NTi Audio reference microphone M2010
Start microphone measurements with the equalized sound source and the reference microphone M2010
Setup for Multiple Microphones with Switcher
This setup verifies several single microphones in a sequence.
Calibrate and equalize the sound source by using the NTi Audio reference measurement microphone M2010
Start microphone measurement sequence with IS-1002 audio input switcher
Turntable Setup for Polar Pattern Measurement
This setup records the polar plot of a single microphone on a turntable in a test stand with equalized speaker.
Calibrate the SPL of the sound source by using the NTi Audio reference measurement microphone M2010
Start automated polar pattern measurement with equalized sound source
Microphone Test Leaflet
프로 오디오 마이크 테스트
copyright 2017 by NTi Audio