XL2 Sound Level Meter
Minirator MR-PRO / MR2
Calibration & Service
Webinars on Demand
Events / Training
My NTi Audio
NTi Audio News
20. Apr. 2015
Octave and one-third octave band measurements with the FX100 Audio Analyzer
Octave and one-third octave band representations of measurements are widely used for real-time acoustic spectral analysis (RTA) in sound level meters. Audio Analyzers, on the other hand, typically represent spectrums with high-resolution FFT's. There are also audio measurement applications where standards require octave or one-third octave band measurements. NTi Audio provides a tool to derive octave and one-third octave band results from FFT measurement results using the FX100 Audio analyzer.
The octave and one-third octave results derived by the FX100 would be applied, for example to:
Sound pressure level evaluation of headphones or earphones (DIN50332)
Sound attenuation measurements in the automotive and building acoustics industries.
Noise emission from appliances and equipment.
How does it work?
In a real-time spectral analysis (RTA), the measurement signal of the microphone is fed continuously through band-pass filters. Each octave or one-third octave band has its own filter. The outputs of the filters are continuously time-averaged and presented as octave / one-third octave bands.
In an FFT, the measurement signal is acquired over a precisely defined time, and transformed into the frequency domain.
For calculation of octave and one-third octave band levels from FFT results, the FFT signal is filtered and summed up. The filters meet the requirements of Class 0 according to the IEC61260 standard.
Consistent Signal Required
The FFT measurement is not continuous. Therefore, in order to convert to octave and one-third octave band the measured signal must produce a constant spectrum with no large dynamic level changes. A common test signal is, for example, Pink noise. Also, noise emission from appliances or machinery (e.g. electric or internal combustion engine) is suitable.
Octave and one-third octave band measurements in FX Control
Contained in the demo projects of the FX Control V2.8.0 software, is an application example for octave and one-third octave band measurements. In this example, an FFT is first measured and the high-resolution data written to a file. The application then calculates the octave and one-third octave results from the FFT data. This data is imported back into the FX Control software for presentation or to be used in further calculations. The measurement can also be evaluated with an "A-weighting" filter. The complete process is automated.
The FX-Control Software V2.8.0 and example projects are available for free download to registered customers at
Contact NTi Audio for
Read more about the
FX100 Audio Analyzer
Press Picture(s) for download here.
Subscribe to Newsletter
Your browser does not support inline frames
Handheld Exel Line (75)
Desktop Flexus Line (42)
12. Sep. 2016
Simple Live Sound Monitoring with XL2
The XL2 Sound Level Meter is the ideal instrument for level monitoring of live events. Using pre-defined profiles, the device starts measuring ...
6. Sep. 2016
Uncompromising Quality Control at NTi Audio
NTi Audio develops and manufactures high-quality sound, audio and vibration measurement products. Every new Sound Level Meter is tested prior to ...
5. Jul. 2016
Accountable Acoustic Quality Control in Production Lines
NTi Audio provides a reliable yet cost-effective solution for acoustic quality control of products in a production line. The XL2 Sound Level Meter ...
Events / Training
Security Essen, 2016
27 - 30 September 2016
3 - 4 October 2016
Vortrag "Schallpegelmessung an Musikveranstaltungen"
11. Oktober 2016
Workshops in Singapore
19. or 20. October 2016
Your News Contact
copyright 2016 by NTi Audio